Validating igloo coolers that are used to 'store' blood in the operating room
A blood banker in Massachusetts is writing a policy/procedure for the use of igloo coolers to 'store' blood in the operating room for selected surgical cases. Since this is new to them, they are wondering what others do with regards to validation of the coolers to hold their temperature. The inquiring blood banker wants to know other institutional practice regarding whether each cooler is validated (or just a representative cooler), and whether the validation is done once upon initiating use of a cooler or on a scheduled basis such as semi-annually? The inquiring blood banker also wants to know if others validate each cooler according to the number of units to be 'stored' in them, i.e. one validation for igloo coolers intended to store 2 units versus a separate validation for igloo coolers intended to store 10 units?
The following responses were submitted:
- The e-network might find the information at on a previous forum issue (one of our first postings on this web site, back in 1999!) to be germane to this discussion.
- A blood banker reported that at her hospital they validate each cooler separately upon arrival in the Transfusion Service and annually thereafter. The validation involves placing the appropriate ice/refrigerant packs into the cooler and recording the temperature each hour. The cooler is deemed acceptable if it maintains a temperature of 1-10 deg. C during the entire 8-hour test period. Coolers failing the original test, have the testing repeated. Failure at both attempts results in the cooler being discarded. Each cooler bears a label indicating the date of validation, the techs initials, and the statement "QC acceptable".
- A blood banker in Texas reports that they have the igloo coolers that come with 2 frozen and 1 cold "Brite-Ice 40" inserts and a holder for 1 - 6 units of packed cells. They validate the igloos by placing 6 units of blood in the holder and placing the frozen and cold inserts in place. They place a thermometer between 2 of the units and took the temperature every hour. The temperature went up to 8 deg C at 12 hours and they stopped the validation at that point. They will sign out the cooler for no longer than 12 hours. For larger coolers, they validate by placing the maximum number of blood units allowed in it and proceeding as above. Ten deg C is the temperature at which blood is no longer allowed to be returned to the blood bank for re-issue. They would choose a temperature below that for the cutoff and then set the corresponding time as the end time for the cooler's efficaciousness.
ADDENDA Feb. 15, 2007
- The Editors believe that the discussion with Alan Williams from the 2006 Ask the FDA Session under "Storage and Transportation" (AABB Membership required to access) and the e-Network Forum discussion "Monitoring temperature of blood products transported within the hospital" are germane to this topic.
Submit comments to the e-Network Forum at firstname.lastname@example.org
Ira A. Shulman, MD
CBBS e-Network Forum Senior Editor & Moderator
W. Tait Stevens, MD
CBBS e-Network Forum Editor & Moderator
Elizabeth M. St. Lezin, MD
CBBS e-Network Forum Associate Editor & Moderator
The e-Network Forum is supported in part by the California Blood Bank Society (CBBS) and the American Red Cross Blood Services (ARCBS) and endorses collegial discussion among blood banking and transfusion medicine professionals. However, neither the CBBS nor the ARCBS in any way endorse the specific views and opinions expressed in the forum. The forum is not intended as a substitute for medical or legal advice and the content should not be relied upon for any medical or legal purposes. Readers should make their own determinations as to: (i) what constitutes appropriate medical, technical, and administrative practices, and (ii) how best to comply with laws and regulations relevant to their questions. For the latter, they should consider consulting, as to any medical matters, a qualified physician, and, as to any legal matters, an attorney familiar with related state and federal laws. The user of the forum, by accessing same, assumes all risks arising out of such use and releases CBBS and their respective members, directors, officers and agents from and against any loss, damage, claim or liability arising out of such use of the Forum.